Electrical muscle stimulation (EMS) has been used successfully in HCI to generate force feedback and simple movements both in stationary and mobile settings. However, many natural limb movements require the coordinated actuation of multiple muscles. Off-the-shelf EMS devices are typically limited in their ability to generate fine-grained movements, because they only have a low number of channels and do not provide full control over the EMS parameters. More capable medical devices are not designed for mobile use or still have a lower number of channels and less control than is desirable for HCI research. We present the concept and a prototype of a 20-channel mobile EMS system that offers full control over the EMS parameters. We discuss the requirements of wearable multi-electrode EMS systems and present the design and technical evaluation of our prototype.
On this page we offer technical details, as schematics, PCB layouts and additional pictures of the prototype. For more information refer to our paper: Zap++: A 20-Channel EMS System for Fine-Grained Wearable Force Feedback
Zap++ is a prototype to generate fine-grained EMS feedback in mobile context. It consists of the following components:
The components are exchangeable. Different grids with different layouts can be connected to the switching boards. Additional switching boards or different kind of switching boards could be connected to the signal generation circuit. The Signal generation unit or the grids connected to the switching boards could be either used seperate as single components.
Top view: All SMD parts are placed on this side. On Top left of the picture is the Atmega328p with a quartz. In the center of the picture are the digital potentiometers and the operational amplifier placed for the current regulation. The IC on the right is a serial to USB converter.
Bottom view: A Bluetooth device can be connected to the header on top of the picture. The header on the bottom of the picture connects the switching boards. The Mosfets placed on this side are parts of the H-Bridge. The green connector on the left side is the header for the stimulation voltage of 87V. The other green header is used for powering the board and the microcontroller.
Below are pictures of the schematics of the signal generation board. The left picture shows the control and communication part including the Bluetooth connection and the serial interface for the switching boards. The right schematics shows the H-bridge and current regulation part.
The layout of the board is designed as two sided PCB. But it is also possible to use a single sided PCB. The blue wires in the right figure have to be connected manually.
The switching board consists of a LED driver IC (MAX6969), LEDs for debug purposes and photo MOS relays. A LED glows if the connected relay is activated. The remaining space on the board is used for signal routing. Jumper on the center of the board enable the configuration of 8 electrodes. A group of eight electrodes could be connectred either to clamp A or B on the H-bridge.
The switching boards are connected serialy. It is possible to connect multiple switching boards in row. Except of one signal all signals are routed directly from the input header to the output header. Only the data signal is routed through the MAX6969.
The switching boards are single sided. As this is a first prototype we used through hole components instead of SMD components.
The electrodes we used were carbon rubber electrodes. These were cutted into smaller pieces and then molded in silicon. The pictures below show the produced sleeves and the form, that we created with a 3D Printer.
Video is coming soon!
Zap++ is developed by the Human-Computer Interaction Group at the University of Hannover. For additional details, see our MobileHCI 2017 paper on Zap++: A 20-Channel EMS System for Fine-Grained Wearable Force Feedback (available on our homepage or in the ACM digital library). For inquiries, please contact firstname.lastname@example.org